Where can I get the 1740A+ 1750A or 1760 calibration software?
Click the link below to download the 17xx series calibration software which normally comes on a floppy disk bundled with the service manual for these instruments. Once downloaded, unzip this archive and copy the contents of it to a floppy disk. Then run the CAL.EXE program from that floppy disk on a PC running DOS. From here the software and the 1740A service manual should walk you through the necessary steps for calibrating your instrument.
NOTE: You will need to run this software on a PC that is running true MS-DOS. Simply running the software in a DOS command windows within Windows will not work because the tool needs access to hardware ports which are protected under Windows operating systems.
This FAQ Applies to:
No product series
FAQ ID 62361View all FAQs »